Infant Foot Wrap SpO2 Sensor #72 (under 25 lbs)
Recommended Patients : Infants and toddlers under 25 lbs
Preferred Site : Patient’s foot, on the edge of instep right next to the little toe.
This sensor is only compatible with CMI Health PC-66H/L Handheld Pulse Oximeter series.
The blue wrap should be changed every 2-3 weeks depending on using scenario. Replacement wrap could be found at: http://www.cmihealth.com/product/foot-wrap-replacement-sensor-72/
For more detail of product information, operation and warning, please download the product manual: http://www.cmihealth.com/wp-content/uploads/2018/02/CMI_072_manual_062717_final-1.pdf
SpO2 measuring range: 35%~100%
SpO2 measuring accuracy: Arms value (defined in ISO 9919 /ISO 80601-2-61) is not greater than 3% in the range of 70%~100%.
Pulse Rate measuring range: 30bpm~250bpm
Pulse Rate measuring accuracy: ±2bpm or ±2%,
whichever is greater.
Wavelength: Red light: 663nm, Infrared light: 890nm